Automatic External Defibrillators Recalled
MRL, Inc. a Welch Allyn Company Issues a Voluntary Worldwide Recall of Selected AED20 Automatic External Defibrillators Contact: Tom McCall 1-315-685-3696 / 1-315-857-7248
FOR IMMEDIATE RELEASE -- Buffalo Grove, IL -- May 10, 2005 -- MRL, Inc., a Welch Allyn Company, today announced it is initiating a voluntary worldwide recall of 597 AED20 Automatic External Defibrillators manufactured in Buffalo Grove, IL between February and July of 2004. The AED20 may display a "Defib Comm" error message on the device display during use resulting in a failure of the device to analyze the patient's ECG and deliver the appropriate therapy which prevents the defibrillator from resuscitating a patient. This problem occurs when an impact to the exterior of the AED20 causes a circuit board connector to perforate an insulation shield enabling an electrical short between the connector and the external housing of the AED20.
The company has received 12 related complaints with this specific group of AED20's, including one instance in which the short may have prevented patient resuscitation. This corresponds to 2 percent of the 597 recalled devices. The company has taken corrective action and no other devices besides those manufactured between the above dates are subject to this voluntary recall.